|Spectrum Range:||2 ~ 25 Microns|
|Power Consumption:||≤ 250W|
|Lifetime of Mineral Plate:||1000 ~ 1500 hrs|
|Diameter of Mineral Plate:||6.5″|
|Auto-Timer:||5 ~ 60min|
|Shipping Dimensional Weight:||40 lbs|
1. Insert the plug into a rated voltage power supply socket. To turn on the power, rotate the timer knob clockwise to the desirable time setting. The indicator will be lit and the lamp will be ready for use after being preheated for 10-15 minutes. Note: Never turn the timer knob counterclockwise or force the timer to return to zero. To terminate the power before time is up, disconnect the power supply.
2. Expose the troubled part of the body or the relevant acupuncture point directly toward the lamp head with a distance of 20 to 30 cm (8 to 12 inches). The best result is obtained when the local skin temperature is kept at 40’C (104’F) or the patient feels comfortable.
A too low temperature will diminish the therapeutic effects and a too high temperature tends to burn the skin.
3. Treatment time length: 20 to 60 minutes each time, 1 or 2 times a day.
Note: The emission plate’s effectiveness decreases after 1,000 hours of usage. Replacement of the emission plate after 1,200 to 1,500 hours is recommended.